Today the scales did not go anymore.
The display remained black.
So I had to find a way to disassemble.
Extra rausholen the Heißluftföhn and time consuming to heat the glass plate? It would have to be easier!
No parts required.
Since it is not important in the balance to get the adhesive surfaces take a beveled spatula. It's much easier and faster.
Contrary to the other instructions, the board is located directly on the plastic housing. Through the 3 holes water gets into the balance and through the adhesion it spreads over the complete underside of the board
The Libra continues to change. Only this time too much water had penetrated which had spread under the board.
Clean the underside of the board with acetone. Alternatively, isopropanol.
After the board was dry, the scales spoke to me again. And so that water does not penetrate again, I taped the holes with silicone
Finally, glue the round cap to the glass plate with a little silicone glue. Because even if the board has a silicone protective coating, it would not be good if Wassser came from above on the SMD components.
Special thanks to these translators:
These translators are helping us fix the world! Want to contribute?
Start translating ›
Embed this guide
Choose a size and copy the code below to embed this guide as a small widget on your site / forum.