Fluke 77 Series III Dial Replacement
The instructions to replace the dial of the Fluke 77 Series III Multimeter.
- Author: Kishan Patel
- Time estimate: 5 minutes
- Difficulty: Easy
This guide helps you replace the Fluke 77 Series III Multimeter Dial. The dial is used to turn the multimeter on and off, and to switch between measuring modes.
Embed this guide
Choose a size and copy the code below to embed this guide as a small widget on your site / forum.
Step 1 — Dial ¶
Hold front panel with both hands, with your thumbs on the dial axle. The dial axle is the metal rod that comes out from the middle of the white plastic rotating piece.
Firmly press down on the dial axle to separate the dial from the front panel.
If this is difficult to do with your hands, push the dial axle against something hard, like the edge of a table.
Step 2 ¶
With the dial loosened, pull the white plastic rotating piece until it comes off the dial axle.
Step 3 ¶
Finally, push again on the dial axle with your thumb while at the same time pulling the dial off the front panel with your other hand .